# Second

### The Unit – second, s (hākona)

The unit of time, the second, was at one time considered to be the fraction 1/86400 of the mean solar day.

The exact definition of "mean solar day" was left to the astronomers. However, measurements showed that irregularities in the rotation of the Earth made this an unsatisfactory definition. In order to define the unit of time more precisely, the 11th CGPM (1960, Resolution 9) adopted a definition given by the International Astronomical Union based on the tropical year 1900. Experimental work, however, had already shown that an atomic standard of time, based on a transition between two energy levels of an atom or a molecule, could be realised and reproduced much more accurately. Considering that a very precise definition of the unit of time is indispensable for science and technology, the 13th CGPM (1967/68, Resolution 1) replaced the definition of the second by the following:

*The second is the duration of 9 192 631 770 periods of the radiation corresponding to the transition between the two hyperfine levels of the ground state of the caesium 133 atom.*

It follows that the hyperfine splitting in the ground state of the caesium 133 atom is exactly 9 192 631 770 hertz, (hfs Cs) = 9 192 631 770 Hz.

At its 1997 meeting the CIPM affirmed that:

*This definition refers to a caesium atom at rest at a temperature of 0 K.*

This note was intended to make it clear that the definition of the SI second is based on a caesium atom unperturbed by blackbody radiation; that is, in an environment whose thermodynamic temperature is 0 K. The frequencies of all primary frequency standards should therefore be corrected for the shift due to ambient radiation, as stated at the meeting of the Consultative Committee for Time and Frequency in 1999.

### Technical Capability

The time section of MSL operates three caesium 5071A atomic clocks, which form the basis of MSL's realisation of legal NZ time. This is realised by minutely adjusting the output of one caesium clock using one of our two micro phase steppers (Spectradynamics High-Resolution Offset Generators). Designated UTC(MSL), NZ time is monitored continuously through information we supply to the BIPM Time Department and reported on monthly in their Circular T bulletins. Contact our Time staff

Watch a short video about the second here.(external link)